Click on “Download PDF” for the PDF version or on the title for the HTML version.
If you are not an ASABE member or if your employer has not arranged for access to the full-text, Click here for options.
Soil Structure and Texture Effects on the Precision of Soil Water Content Measurements with a Capacitance- Based Electromagnetic Sensor
Published by the American Society of Agricultural and Biological Engineers, St. Joseph, Michigan www.asabe.orgCitation: Transactions of the ASABE. 63(1): 141-152. (doi: 10.13031/trans.13496) @2020
Authors: Jasreman Singh, Derek M. Heeren, Daran R. Rudnick, Wayne E. Woldt, Geng Bai, Yufeng Ge, Joe D. Luck
Keywords: Calibration, Capacitance, Depletion, Irrigation, Precision, Sensor, Soil water content, Structure, Uncertainty.
Capacitance-based electromagnetic soil moisture sensors were tested in disturbed and undisturbed soils.
The uncertainty in estimation of soil water depth was lower using the undisturbed soil sample calibrations.
The uncertainty in estimation of soil water depletion was lower than the uncertainty in volumetric water content.
Undisturbed calibration of water depletion quantifies water demand with better precision and avoids over-watering.
Abstract. The physical properties of soil, such as structure and texture, can affect the performance of an electromagnetic sensor in measuring soil water content. Historically, calibrations have been performed on repacked samples in the laboratory and on soils in the field, but little research has been done on laboratory calibrations with intact (undisturbed) soil cores. In this study, three replications each of disturbed and undisturbed soil samples were collected from two soil texture classes (Yutan silty clay loam and Fillmore silt loam) at a field site in eastern Nebraska to investigate the effects of soil structure and texture on the precision of a METER Group GS-1 capacitance-based sensor calibration. In addition, GS-1 sensors were installed in the field near the soil collection sites at three depths (0.15, 0.46, and 0.76 m). The soil moisture sensor had higher precision in the undisturbed laboratory setup, as the undisturbed calibration had a better correlation [slope closer to one, R2undisturbed (0.89) > R2disturbed (0.73)] than the disturbed calibrations for the Yutan and Fillmore texture classes, and the root mean square difference using the laboratory calibration (RMSDL) was higher for pooled disturbed samples (0.053 m3 m-3) in comparison to pooled undisturbed samples (0.023 m3 m-3). The uncertainty in determination of volumetric water content (θv) was higher using the factory calibration (RMSDF) in comparison to the laboratory calibration (RMSDL) for the different soil structures and texture classes. In general, the uncertainty in estimation of soil water depth was greater than the uncertainty in estimation of soil water depletion by the sensors installed in the field, and the uncertainties in estimation of depth and depletion were lower using the calibration developed from the undisturbed soil samples. The undisturbed calibration of soil water depletion would determine water demand with better precision and potentially avoid over-watering, offering relief from water shortages. Further investigation of sensor calibration techniques is required to enhance the applicability of soil moisture sensors for efficient irrigation management.(Download PDF) (Export to EndNotes)