Click on “Download PDF” for the PDF version or on the title for the HTML version.
If you are not an ASABE member or if your employer has not arranged for access to the full-text, Click here for options.
Technical Note: Evaluation of a Rhodamine-WT Dye/Glycerin Mixture as a Tracer for Testing Direct Injection Systems for Agricultural Sprayers
Published by the American Society of Agricultural and Biological Engineers, St. Joseph, Michigan www.asabe.orgCitation: Applied Engineering in Agriculture. 28(3): 643-646. (doi: 10.13031/2013.42424) @2012
Authors: J. D. Luck, S. A. Shearer, B. D. Luck, F. A. Payne
Keywords: Pesticide application, Precision agriculture, Spraying equipment, Pesticides
The purpose of this study was to provide valuable insight regarding the use of Rhodamine WT (red) dye as a tracer for evaluating injected concentrations. More specifically, the effects of mixing the dye with glycerin to simulate the viscosity of a pesticide (e.g., glyphosate) or injecting the dye/glycerin mixture into deionized (DI) versus tap water on developing appropriate calibration equations were evaluated. Test results indicated that mixing the dye in a solution of glycerin and DI water significantly affected absorbance measurements compared to the dye mixed solely in DI water. The error in estimating absorbance was 7.4% between the two calibration equations. Therefore, any calibration curves must include a solution containing glycerin to compensate for this. Absorbance results also indicated that potable tap water could be used to simulate the spray carrier as opposed to the DI water. A calibration curve was developed for the simulated pesticide (dye/glycerin/DI water solution) injected into the carrier (tap water) for solutions ranging from 2,000:1 (carrier:pesticide) to 500,000:1; an overall dilution range of 250:1. This dilution range exceeded typical pesticide tank-mixed dilutions which were on the order of 11:1 for the application of glyphosate for corn or soybeans. The regression model standard error for predicting the dye concentration based on absorbance measurements was 5.3x10-6.(Download PDF) (Export to EndNotes)